Standard Test Method for the Continuous Measurement of Dissolved Ozone in Low Conductivity WaterName übersetzen
NORM herausgegeben am 15.7.2016
Bezeichnung normen: ASTM D7677-16
Ausgabedatum normen: 15.7.2016
Zahl der Seiten: 5
Gewicht ca.: 15 g (0.03 Pfund)
Land: Amerikanische technische Norm
Kategorie: Technische Normen ASTM
dissolved gas, high purity water, ozone, pharmaceutical water, sanitization,, ICS Number Code 17.120.01 (Measurement of fluid flow in general)
|Significance and Use|
5.1 Dissolved Ozone is useful in many industries for water sanitization, TOC reduction, food preservation, cleaning-in-place of food and beverage systems, and pyrogen destruction. It is often necessary to know how much ozone has entered the water, how much remains, and the degree to which it has been removed before process use.
5.2 Some applications require that contact time, DO3 concentration integrated over time, be calculated, to assure disinfection.
5.3 Continuous observation of trends in these measurements are needed for continuous quality monitoring and the measurement may be used for closed loop control of ozonation.
5.4 In many pure water applications and especially where water quality is regulated by the FDA or similar enforcement agencies, ozone removal must be complete before the water is used. This test method is useful for detecting and determining dissolved ozone levels in water at the trace level as well as at process concentrations where sanitization and chemical reactions occur.
1.1 This test method covers the on-line and in-line determination of dissolved ozone (DO3) in low conductivity water in the range from 0.001 mg/L to 5.0 mg/L DO3 and conductivity <100 μS/cm, typical of pharmaceutical and microelectronics pure waters. DO1.2 This test method provides a more convenient means for continuous measurement than the colorimetric methods typically used for grab sample measurements.
1.3 This test method has the advantage of high sensitivity as well as durability in the process environment and has few interferences.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|2. Referenced Documents|
Wollen Sie sich sicher sein, dass Sie nur die gültigen technischen Normen verwenden?
Wir bieten Ihnen eine Lösung, die Ihnen eine Monatsübersicht über die Aktualität der von Ihnen angewandten Normen sicher stellt.
Brauchen Sie mehr Informationen? Sehen Sie sich diese Seite an.
Letzte Aktualisierung: 2021-09-17 (Zahl der Positionen: 2 524 940)
© Copyright 2021 NORMSERVIS s.r.o.