Standard Test Method for Spectral Responsivity Measurements of Photovoltaic DevicesName übersetzen
NORM herausgegeben am 1.4.2019
Bezeichnung normen: ASTM E1021-15(2019)
Ausgabedatum normen: 1.4.2019
Zahl der Seiten: 11
Gewicht ca.: 33 g (0.07 Pfund)
Land: Amerikanische technische Norm
Kategorie: Technische Normen ASTM
cell, irradiance, measurement, photovoltaic, quantum efficiency, radiant power, responsivity, solar, spectral, testing ,, ICS Number Code 27.160 (Solar energy engineering)
|Significance and Use|
5.1 The spectral responsivity of a photovoltaic device is necessary for computing spectral mismatch parameter (see Test Method ). Spectral mismatch is used in Test Method to measure the performance of photovoltaic cells in simulated sunlight, in Test Methods to measure the performance of photovoltaic modules and arrays, in Test Method to calibrate photovoltaic primary reference cells using a tabular spectrum, and in Test Method to calibrate photovoltaic secondary reference cells. The spectral mismatch parameter can be computed using absolute or relative spectral responsivity data.
5.2 This test method measures the differential spectral responsivity of a photovoltaic device. The procedure requires the use of white-light bias to enable the user to evaluate the dependence of the differential spectral responsivity on the intensity of light reaching the device. When such dependence exists, the overall spectral responsivity should be equivalent to the differential spectral responsivity at a light bias level somewhere between zero and the intended operating conditions of the device. Depending on the linearity response of the DUT over the intensity range up to the intended operating conditions, it may not be necessary to set up a very high light bias level.
5.3 The spectral responsivity of a photovoltaic device is useful for understanding device performance and material characteristics.
5.4 The procedure described herein is appropriate for use in either research and development applications or in product quality control by manufacturers.
5.5 The reference photodetector’s calibration must be traceable to SI units through a National Institute of Standards and Technology (NIST) spectral responsivity scale or other relevant radiometric scale. ,5.6 This test method does not address issues of sample stability.
5.7 Using results obtained by this test method and additional measurements including reflectance versus wavelength, one can compute the internal quantum efficiency of a device. These measurements are beyond the scope of this test method.
5.8 This test method is intended for use with a single-junction photovoltaic cell. It can also be used to measure the spectral responsivity of a single junction within a series-connected, multiple-junction photovoltaic device if electrical contact can be made to the individual junction(s) of interest.
5.9 With additional procedures (see Test Methods ), one can determine the spectral responsivity of individual junctions within series-connected, multiple-junction, photovoltaic devices when electrical contact can only be made to the entire device’s two terminals.
5.10 Using forward biasing techniques1.1 This test method is to be used to determine either the absolute or relative spectral responsivity response of a single-junction photovoltaic device.
1.2 Because quantum efficiency is directly related to spectral responsivity, this test method may be used to determine the quantum efficiency of a single-junction photovoltaic device (see ).
1.3 This test method requires the use of a bias light.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Wollen Sie sich sicher sein, dass Sie nur die gültigen technischen Normen verwenden?
Wir bieten Ihnen eine Lösung, die Ihnen eine Monatsübersicht über die Aktualität der von Ihnen angewandten Normen sicher stellt.
Brauchen Sie mehr Informationen? Sehen Sie sich diese Seite an.
Letzte Aktualisierung: 2021-09-19 (Zahl der Positionen: 2 525 031)
© Copyright 2021 NORMSERVIS s.r.o.