Standard Practice for Describing and Specifying Inductively Coupled Plasma Atomic Emission SpectrometersName übersetzen
NORM herausgegeben am 1.11.2016
Bezeichnung normen: ASTM E1479-16
Ausgabedatum normen: 1.11.2016
Zahl der Seiten: 15
Gewicht ca.: 45 g (0.10 Pfund)
Land: Amerikanische technische Norm
Kategorie: Technische Normen ASTM
ICP-AES, optical emission, spectrometry, atomic emission, spectroscopy, emission, spectroscopy, inductively coupled plasma,, ICS Number Code 17.180.30 (Optical measuring instruments)
|Significance and Use|
5.1 This practice describes the essential components of an ICP-AES. The components include excitation/radio-frequency generators, sample introduction systems, spectrometers, detectors, and signal processing and displays. This description allows the user or potential user to gain a cursory understanding of an ICP-AES system. This practice also provides a means for comparing and evaluating various systems, as well as understanding the capabilities and limitations of each instrument.
5.2 Training—The manufacturer should provide training in safety, basic theory of ICP-AES analysis, operations of hardware and software, and routine maintenance for at least one operator. Training ideally should consist of the basic operation of the instrument at the time of installation, followed by an in-depth course one or two months later. Advanced courses are also offered at several of the important spectroscopy meetings that occur throughout the year as well as by independent training institutes. Several independent consultants are available who can provide training, sometimes at the user's site.
1.1 This practice describes the components of an inductively coupled plasma atomic emission spectrometer (ICP-AES) that are basic to its operation and to the quality of its performance. This practice identifies critical factors affecting accuracy, precision, and sensitivity. It is not the intent of this practice to specify component tolerances or performance criteria, since these are unique for each instrument. A prospective user should consult with the manufacturer before placing an order, to design a testing protocol that demonstrates the instrument meets all anticipated needs.
1.2 The values stated in SI units are to be regarded as standard. The values given in parentheses are for information only.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Specific safety hazard statements are given in Section .
|2. Referenced Documents|
Bereitstellung von aktuellen Informationen über legislative Vorschriften in der Sammlung der Gesetze bis zum Jahr 1945.
Aktualisierung 2x pro Monat!
Brauchen Sie mehr Informationen? Sehen Sie sich diese Seite an.
Letzte Aktualisierung: 2020-08-10 (Zahl der Positionen: 2 390 550)
© Copyright 2020 NORMSERVIS s.r.o.